Section I. GENERAL
Restorer, Pulse Form TD-206/G is issued in two configurations which in-
terface with two different cable types. Restorer TD-206B/G is similar to
Restorer TD-206/G except for end cap configuration. Differences are sum-
marized as follows:
1-2. Index of Technical Publications
Restorer, Pulse Form
Special Purpose Electrical
Refer to the latest issue of DA Pam 310-4 to deter-
TD-206/G with 7A2 end cap . . . . . . . . . . . . . . . . . . . . CX-4245/G
mine whether there are new editions, changes or ad-
TD-206/G with 7A2 end cap . . . . . . . . . . . . . . . . . . . CX-11230/G
ditional publications pertaining to the equipment.
TD-206B/G . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . CX-l1230/G
In this manual, the designation TD-206/G applies
1-3. Maintenance Forms, Records, and Reports
to all sets unless otherwise specified. Cable, Special
a. Reports of Maintenance and Unsatisfactory
Purpose Electrical CX-4245/G is similar to Cable
Equipment. Department of the Army forms and
CX-11230/G except for connector configuration. In
procedures used for equipment maintenance will be
this manual, the designation CX-4245/G applies to
those prescribed by TM 38-750, The Army Main-
both cables unless otherwise specified.
b. Report of Packaging and Handling Deficien-
cies. Fill out and forward SF 364 (Report of
This manual describes Test Set, Telephone
Discrepancy R O D ) ) a s p r e s c r i b e d i n A R
stallation, operation, and organizational main-
4355.73/AFR 400-54/MCO 4430.3E.
c. Discrepancy in Shipment Report (DISREP) (SF
1-1.1 Test Set Simplified Instruction Cards
361). Fill out and forward Discrepancy in Shipment
Report (DISREP) (SF 361) as prescribed in AR
When using the Test Set Simplified Instruction Car-
55-38/NAVSUPINST 4610.33B/AFR 75-18/MCO
ds, the following corrections are required:
a. Page 1, METER SELECT Switch Indications.
(1) PCM IN: Meter Indicates that a pcm input
1-3.1. Reporting Equipment Improvement
signal is present.
R e c o m m e n d a t i o n s (EIR)
(2) PCM OUT: Meter indicates that a pcm out-
If your equipment needs improvement, let us know.
put signal is being transmitted.
Send us an EIR. You, the user, are the only one who
b. Page 3, Cable and Restorer Tests.
can tell us what you don't like about your equip-
(1) Paragraph 3F. Place METER SELECT
ment. Let us know why you don't like the design.
switch to DC AMPS. Meter should indicate between
Tell us why a procedure is hard to perform. Put it on
7 and 10.
an SF 368 (Quality Deficiency Report). Mail it to
(2) Paragraph 3G. Place METER SELECT
switch to VOLT DROP position. Meter should in-
Electronics Command, ATTN: DRSEL-ME-MQ,
dicate between 7 and 9 in the green area.
Fort Monmouth, NJ 07703. We'll send you a reply.
(3) Paragraph 3H. Place METER SELECT
switch to PCM IN position. If meter indicates zero,
1-3.2. Administrative Storage
no pcm input signal is being received.
Administrative Storage of Equipment issued to and
(4) Paragraph 3I. Place METER SELECT swit-
used by Army activities will have preventive main-
ch to PCM OUT position. If meter indicates zero, no
tenance performed in accordance with the PMCS
pcm signal is being transmitted.
Change 5 1-1